/ Home / Products / Miscellaneous / Accessories Filters, Regulators and Lubricators Filters, Regulators and Lubricators
For Pneumatic systems, liquid, nylon, and particle filters, air pressure, and air filter regulators are available. Replaceable filter elements remove particles and droplets as small as .01 micron with 93-95% efficiency. Applications are for substantial flow capacity.
How often should I replace the filter on my 1205A-0 handheld CO analyzer?
This filter should be replaced if it begins to appear wet or dirty and can be replaced with Model F195 Air Filter.
What might be the cause for Magnehelic® gages filling approximately 1/3 full of water, with lots of liquid droplets on the glass. Both are outside mounted on the side of an air stripping tower. One is a differential pressure gage that measures the pressure drop across a pitot tube. The other just measures the back pressure in the stripper. What can be done to fix the malfunction?
It looks like you’re getting some condensation in the Magnehelic®, which is causing it to collect on the bottom of the face. Since the Magnehelic® is not encapsulated, it is not recommended to be used in applications where moisture can form. One option to fix the problem is installing a liquid filter (like an F222) to filter out that moisture. Another option is to install a Capsuhelic (Series 4000), which is made to handle both air and liquid.
T: +1 800.872.9141 +1 219.879.8000 F: +1 219.872.9057
Copyright © Dwyer Instruments, Inc. All Rights Reserved.